Laboratory TD4A-WS Desktop Low Speed Centrifuge with 12 holes
Centrifugal separation technology is based on particle in a practical centrifugal occasions and developed to the state of the new technology. Different density, size or shape particles in different occasions centrifugal sedimentation, so a roughly spherical is not uniform mixture, can use the method of centrifugal are isolated, centrifuge is to further research on biological chemistry, separate a lot of material. For example collect cells, separating the plasma, from these the purification of separation in preparations DNA and protein molecules, and can be isolated the virus and massive escherichia coli, strict cell components, such as the ribonucleoprotein complex particle.yf
The machine adopts brushless motor, the microcomputer control, with RCF automatic computing and setting. With a variety of program protection, easy operation etc.
Safety control: the lid lock system in the operation;
In the unbalanced rotating automatic braking system;
Automatic door opened the switch at the end of the operation
Main technical parameters:
1. Use power supply: ac 220 V ± 10 V frequency 50 HZ
2. The whole machine power: 250 W
3. Turn speed: 0-4000 RPM ± 50 turn/points digital display
4. Centrifugal capacity: 8by 15 ml (4000 RPM)
5: relative centrifugal force: 1880 xg
6: envelop dimensions:23 x 290 x 270 (mm)
7: set when: 1-99 minutes digital display
8: heavy quantity: 10 kg
9: more than 65 db noise maximum speed
Three use safety guidelines
1. Use this product to remember that comply with the following safety precautions, read all instructions.
2. Users of the electrical outlet of rated parameters should be not less than the electrical machine is rated parameters and good grounding measures.
3. Replacing the fuse, should will plug from the socket drew.
4. Instruments after failure, there should be a professional repair, use manufacturers not recommended accessories can cause certain quality problems.
5. Have a children's environment use, should close attention.
6. From the heart in operation, it is strictly prohibited to play boot lid, must wait for rotor completely stop turning. To play out on centrifugal cover of the tube.
7. Instruments in use process move is strictly prohibited.
8. It is strictly prohibited to all kinds of liquid or other material into the centrifuge indoor job, otherwise, it will damage the apparatus.
9. The instrument in use, must be supervised.
10. Scrub machine, should first pulled out the power supply.
11. The instrument shell should be properly grounded, lest the moisture influence the accidents.
Method of use:
1. Will join to the centrifugal separation material tube inserted into the machine for respectively the centrifugal tube, in order to make instruments work stable performance is good, avoid to produce larger vibration, each centrifugal tube material must be added basic equal, symmetrical insert the machine for centrifugal tube, close the cover plate.
2. The switch on the power, turn on the power switch (instrument back).
3. With keys to choose a good speed and need to work time.
4. Then press the start button, this time the centrifuge, it started to work. When you get to the set of working time will be issued a warning sound.
1. The right to use and pay attention to the maintenance of the equipment, make its are in good condition, can prolong the service life of the instrument.
2. Instruments in continuous working period, should be three months should be a regular inspection: check if there is water, dirt and fall into the motor and control component inside, check the fuse, control component and tighten the screws.
3. The instrument for long-term use, natural wear is a normal phenomenon, the instrument in the use of a year later, if discover the motors have not normal noise, motor bearing wear, electronic component failure failure, the enterprise will continue to provide high quality service, assist processing.